The Aquamax KF has two different delay times which are operator programmable. Start Delay enables additional time to be added before electrolysis current is applied and the titration starts to count. This is useful if the sample is very viscous or is slow to release water Minimum Titration Time allows the titration to commence and to count as normal but does not enable the end point to be reached until at least the programmed minimum time has elapsed. This is useful when using a vaporiser or sometimes when analysing gas samples which have very low water content.
Posted in: Practical advice